Page Operation gasoline or anaesthetics. Oxygenation in the vicinity of the defibrillation elec- trodes must be strictly avoided. Danger of electric shock! The FRED easyport is a high-voltage electric therapy device. Improper use of the device can endanger life.
|Published (Last):||3 June 2018|
|PDF File Size:||19.69 Mb|
|ePub File Size:||15.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Operation gasoline or anaesthetics. Oxygenation in the vicinity of the defibrillation elec- trodes must be strictly avoided. Danger of electric shock! The FRED easyport is a high-voltage electric therapy device. Improper use of the device can endanger life. Always follow the instruc- tions given in this user guide. Page Ensuring Operational Readiness Inspection Report on page If you only carry out the self-tests, the battery will last for approx.
Other- wise the lives of the patient, the user and bystanders are in danger. The patient must: — not come into contact with other persons during defibrillation. Please note that the indicated expiration date only applies if the vacuum pack is intact. The pads are pre-gelled, so there is no need to use extra contact agent.
Proceed as follows: 1. Exactly follow the instructions. Step 1 Switching on and preparing the device Switch the device on by pressing the green button 1. Page 24 Defibrillating FRED easyport Defibrillation Procedure Step 3 Shock delivery As soon as the energy for a shock is charged, the device prompts the user to deliver the shock by pressing button 3.
Before you release the shock, check the displayed ECG curve to make sure that a shockable rhythm is present. Switch the device off as soon as the therapy is finished keep the button pressed for approx. Disconnect the plug of the electrode line. See section Error Detection 6.
The test results must be recorded and compared to the values in the accompanying documents. The following table indicates the intervals and responsibilities of the maintenance work required.
믿을 수 있는 간편결제! 스마일페이